- Open Access
Ventriculoperitoneal shunt complications: a local study at Qena University Hospital: a retrospective study
Egyptian Journal of Neurosurgery volume 33, Article number: 8 (2018)
Shunting of cerebrospinal fluid (CSF) has reduced the morbidity and mortality of hydrocephalus. Ventriculoperitoneal (VP) shunt is the most commonly used procedure for shunting, but it has potential complications that may need multiple surgical interventions.
It is a clinical retrospective observational study that was conducted on 30 patients of both genders with different ages who presented with signs and symptoms of shunt malfunction between 2012 and 2016. A complete clinical assessment was done, a final diagnosis was made, and a treatment of individual patients was planned accordingly. Patients were followed up within 1 week and 1 month post-operatively. Data was analyzed using the SPSS (version 16.0).
VP shunt was inserted for 205 patients. Thirty (14.6%) patients had various forms of complications. Fifteen (50%) patients had complications related to the proximal catheter and the reservoir while 15 (50%) patients had complications related to distal catheter. The most common complications were exposure of the shunt 23.3% (13.3% exposed shunt reservoir and 10% exposed distal catheter) followed by shunt obstruction 13.3% (6.66% proximal and 6.66% distal). Twenty-eight (93.3%) patients were managed surgically, 24 (85%) patients of them showed marked improvement at the end of the first month postoperatively, while four (15%) patients needed another surgical intervention.
Insertion of V-P shunt is routinely done by all neurosurgeons. A great care should be taken during insertion of the shunt system starting from scrubbing to avoid complications. Despite complications, the VP shunt remains the main surgical procedure used for hydrocephalus management.
Hydrocephalus is an excess of cerebrospinal fluid (CSF) in the ventricular system due to the imbalance between formation and absorption of CSF which is referred to (i) obstruction of the CSF pathways, (ii) over production of CSF, and (iii) impaired venous drainage . Hydrocephalus is the second most common congenital brain malformation . To date, the standard treatment of hydrocephalus is ventriculoperitoneal shunt. The technique of using the peritoneal cavity for CSF absorption in ventriculoperitoneal shunting (VPS) was developed by Kausch in 1908 . Although VPS insertion is a common neurosurgical procedure, complication rates in adults are poorly established with a reported range from 17 to 33% [2, 4,5,6,7,8,9,10]. Children demonstrated a higher rate of shunt complications than did adults at 5 years (48 versus 27%, P < 0.0001) . The advent of endoscopic third ventriculostomy has gained popularity due to the high complication and failure rates of ventriculoperitoneal shunt . The major disadvantage of VPS is the fact that it constitutes a foreign body and prone to complications such as mechanical blockage, shunt infection, shunt migration, and rarely shunt protrusion . If a shunt system fails to be operated correctly, the patient’s life and cognitive functions are placed at risk. Thus, an urgent revision must be done . Although developing countries face the problem of shunt complications more than other countries, much research is ongoing but still remains a common problem .
This is a retrospective observational study that was conducted at the Department of Neurosurgery, Qena University Hospital, South Valley University for a period of 5 years from January 2012 to December 2016. An informed signed consent was taken from the parents and guardians of patients before enrolling them into the study after the approval of the ethical committee of the Faculty of Medicine, South Valley University. During the study period, a complete clinical assessment including a detailed history and examination with a particular emphasis on neurological examination was done for all patients after admission. A medium-pressure PS medical valve systems were inserted for all patients. Patients with VP shunt complications operated at Qena University Hospital were included while patients operated before the study period were excluded. The investigations performed for all patients were complete blood count (CBC), erythrocyte sedimentation rate (ESR), complete urine analysis, X-ray chest, and brain computed tomography (CT) scan or MRI. Specific investigations such as CSF analysis, CSF culture and sensitivity, blood culture and sensitivity, urine culture and sensitivity, pus culture and sensitivity, ultrasound of abdomen, shunt series X-rays, and MRI of the brain were also performed when indicated. A final diagnosis was made on the basis of clinical findings, and investigations and treatment of individual patients were planned accordingly. All patients with VP shunts who had one or more complications were included in the study.
Data was recorded and analyzed using the Statistical Package of Social Sciences (SPSS) version 16. Descriptive statistics were presented as frequencies, percentages, means, and standard deviations.
Two hundred and five patients for whom VP shunt was inserted by consultant and senior neurosurgeons, thirty (14.6%) patients had complications. Both genders were included [22 (73.3%) males and 8 (26.6%) females]. All ages were also included [22 (73.3%) infants, 6 (20%) children, and 2 (6.6%) adults] with a mean age (3.60 ± 6.64) as shown in Table 1. Clinically, 22 (73.3%) patients presented with bulged anterior fontanelle, 21 (70%) with sleepiness, 20 (66.7%) patients with frequent vomiting, 17 (56.66%) patients with enlarged head, 12 (40%) patients with tense anterior fontanelle, 11 (36.7%) patients with agitations and irritability, nine (30%) patients with inflammation around distal tube, seven (23.33%) patients with headache, six (20%) patients with fever, four (13.66%) patients with exposure of the shunt reservoir, three (10%) patients with CSF leak from the distal wound, two (6.66%) patients with swelling around the reservoir, two (6.66%) patients with swelling around distal tube, and one (3.33%) patient with scrotal swelling as shown in Fig. 1 and Table 1.
The documented complications were classified according to site into two categories: (i) complications related to proximal catheter and reservoir and (ii) complications related to distal catheter as presented in Table 1. Concerning proximal catheter and reservoir complications, they appeared in 15 patients. The skin erosion over the shunt reservoir was the most common complication which was documented in four (13.3%) patients as shown in Fig. 2. Misplaced catheter was noted in three (10%) patients; ventriculitis was presented in two (6.7%) patients as shown in Fig. 3. Obstruction of proximal catheter by debris was noted in two (6.7%) patients, and CSF leak making collection around the reservoir was documented in two (6.7%) patients as shown in Fig. 4. Overdrainage of the ventricles may lead to unilateral or bilateral chronic subdural hematoma (CSDH), and this was presented in two (6.7%) patients as shown in Fig. 5. Regarding distal catheter complications, they also occurred in15 patients. Exposure of skin overlying the tube occurred in three (10%) patients (hyperemia with superficial ulceration or complete exposure) as shown in Fig. 6. Infection around the distal catheter was reported in two (6.7%) patients. Two (6.66%) patients showed poor peritoneal absorption which leaded to distal failure and CSF collection under the skin surrounding the distal catheter as shown in Fig. 7. Fracture can occur at any site along the course of the distal tube especially near bony prominences. In our study, two (6.66%) patients were documented to have a fracture and, hence, distal tube migration as shown in Fig. 8. Extrusion of the distal end of distal catheter through anus was also reported in two (6.66%) patients as shown in Fig. 9. Two (6.66%) patients had a distal failure due to obstruction of the distal end of the distal catheter by pseudo cyst causing closed narrow space for CSF drainage causing VP shunt malfunctioning as shown in Fig. 10. Other complications were noted as extraperitoneal placement of the distal catheter and hydrocele, each of them noted in only one patient for each (3.33%). It was noted that complications tend to occur more frequently in case of proximal catheter insertion in occipital horn (posterior parietal shunt) than in frontal horn (frontal VP shunts). Such complications occurred within 26 out of 162 patients for whom occipitoparietal VP shunt was done. The remaining 43 patients for whom frontal VP shunt was done showed complications related to four patients only. Thus, the incidence of complications with occipitoparietal VP shunt is 86.67 and 13.33% with the frontal type out of the total number of complications. VP shunts were inserted for treatment of hydrocephalus caused by any etiology such as congenital, post meningitic, post subarachnoid hemorrhage, and tumor obstructing CSF pathway. In the current study, out of 205 patients operated with VP shunt, 188 patients were congenital, 10 patients were post meningitic, four patients were post subarachnoid hemorrhage, and three patients were tumor-induced hydrocephalus. Out of 30 patients who had various types of VP shunt complications, 28 (93.33%) patients complained of congenital hydrocephalus, one (3.33%) patient complained of post meningitic hydrocephalus, and one (3.33%) patient complained of post subarachnoid hydrocephalus. No patient who had tumor-induced hydrocephalus complained of shunt complications (see Table 2).
Patients were followed up after 1 week and 1 month postoperatively. Two (6.66%) patients were managed conservatively while 28 (93.3%) patients were managed surgically. Surgeries included proximal revision, distal revision, debridement, redirection of misdirected catheter, repositioning of the distal catheter, new VP shunt on the other side, changing the shunt pressure to higher one in case of CSDH, or closure of hernia sac as shown in Fig. 11. At the end of the first month, two (6.66%) patients with infection around the distal tube were managed conservatively by good antiseptics and highly sensitive antibiotics according to culture and sensitivity tests showing marked improvement. Out of the 28 surgically treated patients, 24 (85%) patients showed marked improvement while four (15%) patients needed other surgical interventions.
Although VP shunt is an effective treatment of hydrocephalus, it is plagued by shunt-related complications . As VP shunt is lifelong commitment, multiple surgical procedures may be required during life time . The incidence of VP shunt complications was reported by most studies to be slightly higher in males than females , which was consistent with the current study as males represent 73.3%. Among 30 patients who were operated by ventriculoperitoneal shunt, infants and children represented 93.3% whereas adults represented 6.6%. This is consistent with Abdul Munam et al. who conducted their study on 40 VP shunt complicated patients where children represented 85% . In neonates, scalp necrosis is actually a common complication associated with VP shunts which is due to the inherent skin fragility and the superficial nature of the shunt [20, 21]. In agreement with this current study scalp, necrosis was presented in four (13.3%) patients. Lee et al. found shunt blockage in 12.2% of 246 shunt procedures in Seoul, Korea, and their infection rate was 4.1%. Shunt infection was found together with blockage in most instances in their series indicating that shunt malfunction could have been caused by infection in these patients . Vanaclocha et al. observed that shunt malfunction occurred in infected shunts where some of which were clinically undetectable. They argued that the incidence of shunt infection might be higher than generally reported and that negative cultures of CSF taps did not exclude shunt infection in malfunctioning shunts . Peacock and Currer found shunt blockage to be 20% in their series of 440 patients . Mwan’gombe and Omulo reported an infection rate of 24.6% among children operated for non-tumor hydrocephalus in Nairobi . In the current study, shunt obstruction was reported in approximately 13.4% of patients (6.7% proximal obstruction by debris and 6.7% distal obstruction by pseudo cyst). Infection was also reported in 13.4% of patients. Hamada and Abou Zeid found that misdirection of proximal catheter was founded in two (7.1%) patients of their shunt malfunction series which is approximately near to the result of this current study as proximal catheter misdirection was founded in three (10%) patients . Aldrich and Harmann found that shunt disconnection and fracture accounted for 15% of their shunt malfunctions and that occipitally placed shunts had a higher tendency to dislocate than frontally placed shunts . In agreement with this study, Shunt fracture was noted in 6.66% of shunt complication for occipitally placed shunts. Bierbauer et al. found no advantage of anteriorly placed shunts over posteriorly placed shunts in terms of shunt malfunction or infection . However, in the current study, there was an advantage of anteriorly placed VP shunts over posteriorly placed shunts in terms of malfunction and infection. It was also noted that complications tend to occur more with occipitoparietal than with frontal VP shunts. In more details, incidence of complications with occipitoparietal VP shunt was 10.4% and with frontal type was 6%.
Abdominal complications of VP shunt are not rare and the main causes of distal catheter failure are related to extra peritoneal retraction of the catheter and subcutaneous or intra-abdominal cerebrospinal fluid (CSF) collections . In the current study, intra-abdominal pseudo cyst was reported in two (6.66%) patients and extra peritoneal catheter in one (3.33%) patient.
A higher incidence of unobliterated processus vaginalis in pediatric patients than in adult patients leads to a higher likelihood of VP shunt distal catheter migration into the scrotum . In the current study, one infant presented with scrotal swelling due to patent processus vaginalis (see Fig. 11) and this is consistent with the previous study. Sathyanarayana et al. documented a protrusion of distal catheter per anus without any complications such as obstruction or peritonitis . This agrees with the current study where two patients were reported with a VP shunt distal catheter per rectum without any complications in the form of obstruction or peritonitis. Previous studies reported that age and principal diagnosis (etiology) are independent contributors to the risk of initial shunt failure [32,33,34]. In agreement with these studies, the current study demonstrated that age and etiology were significantly associated with shunt revision where 28 (93.33%) patients with VP shunt complications were congenital. Accordingly, there was a higher rate of complications related to congenital etiology.
Insertion of VP shunt is routinely done by all neurosurgeons. Great care and precautions should be taken during insertion of the shunt system starting from scrubbing to avoid complications such as infection. Implantation of VP shunt system should be done by well-qualified neurosurgeons in order to limit or avoid shunt complications. Despite complications, the VP shunt remains the main surgical procedure used for hydrocephalus management.
Complete blood count
Chronic subdural hematoma
Erythrocyte sedimentation rate
Magnetic resonance imaging
Statistical Package of Social Sciences
Kandasamy J, Jenkinson MD, Mallucci CL. Contemporary management and recent advances in paediatric hydrocephalus. BMJ. 2011;343:d 4191.
Wu Y, Green NL, Wrensch MR, et al. Ventriculoperitoneal shunt complications in California: 1990 to 2000. Neurosurgery. 2007;61:557–62.
Chung JJ, Yu JS, Kim JH, et al. Intraabdominal complications secondary to ventriculoperitoneal shunts: CT findings and review of the literature. AJR Am J Roentgenol. 2009;193:1311–7.
Korinek AM, Fulla L, Boch AL, et al. Morbidity of ventricular cerebrospinal fluid shunt surgery in adults: an 8-year study. Neurosurgery. 2011;68:985–94. discus-sion: 994-995
Puca A, Anile C, Maira G, et al. Cerebrospinal fluid shunting for hydrocephalus in the adult: factors related to shunt revision. Neurosurgery. 1991;29:822–6.
Khan F, Rehman A, Shamim MS, et al. Factors affecting ventriculoperitoneal shunt survival in adult patients. Surg Neurol Int. 2015;6:25.
Reddy GK, Bollam P, Caldito G. Long-termout-comes of ventriculoperitoneal shunt surgery in patients with hydrocephalus. World Neurosurg. 2014;81:404–10.
Reddy GK. Ventriculoperitoneal shunt surgery and the incidence of shunt revision in adult patients with hemorrhage-related hydrocephalus. Clin Neurol Neurosurg. 2012;114:1211–6.
Reddy GK, Bollam P, Caldito G. Ventriculoperitoneal shunt surgery and the risk of shunt infection in patients with hydrocephalus: long-term single institution experience. World Neuro-surg. 2012;78:155–63.
O’Kelly CJ, Kulkarni AV, Austin PC, et al. Shunt-dependent hydrocephalus after aneurysmal subarachnoid hemorrhage: incidence, predictors, and revision rates. Clinical article. J Neurosurg. 2009;111:1029–35.
Yvonne W, Nella L, Green, et al. Ventriculoperitoneal shunt complications in California. Neurosurgery. 2007;61(3):557–63.
Bouras T, Sgouros S. Complications of endoscopic third ventriculostomy: a systematic review. Acta Neurochir Suppl. 2012;113:149–53.
Ribaupierre S, Rilliet B, Vernet O, et al. Third ventriculostomy vs ventriculoperitoneal shunt in pediatric obstructive hydrocephalus: results from a Swiss series and literature review. Childs Nerv Syst. 2007;23(5):527–33. https://doi.org/10.1007/s00381-006-0283-4.
Omotayo A, Olumide E, Okezie O, et al. Unusual complication of ventriculoperitoneal shunt. Romanian Neurosurg. 2013;XX 4:375–8.
Mubarak H, Riaz A, Aleem-ud-Din Sh, et al. Ventriculoperitoneal shunt blockage. J Ayub Med Coll Abbottabad. 2012;24 (3-4):82–4.
Shao Y, Li M, Sun JL, et al. A laparoscopic approach to ventriculoperitoneal shunt placement with a novel fixation method for distal shunt catheter in the treatment of hydrocephalus. MinimInvasive Neurosurg. 2011;54(1):44–7.
Reddy GK. Ventriculoperitoneal shunt surgery and the incidence of shunt revision in adult patient with hemorrhage-related hydrocephalus. Clin Neurol Neurosurg. 2012;114(9):1211–6.
Ghritlaharey RK, Budhwani KS, Shrivastava DK, et al. Ventriculoperitoneal shunt complications needing shunt revision in children: a review of 5 years of experience with 48 revisions. Afr J Paediatr Surg. 2012;9(1):32–9.
Abdul Munam, Vashdev, Riaz A. Pattern of complications and presenting features in patients implanted ventriculoperitoneal shunt due to hydrocephalus JLUMHS 2014;13(02):57.
Ammar A, Nasser M. A long-term complication of burying a shunt valve in the skull. Neurosurg Rev. 1995;18:65–7. PMID: 7566533
Bot GM, Ismail NJ, Usman B, et al. Subpericranial shunt valve placement: a technique in patients with friable skin. Childs Nerv Syst. 2014;30:1431–3. PMID: 24839037
Lee JY, Wang KC, Cho BK. Functioning periods and complications of 246 cerebrospinal fluid shunting procedures in 208 children. J Korean Med Sci. 1995 Aug;10(4):275–80.
Vanachola V, Zais Sapena N, Leiva J. Shunt malfunction in relation to shunt infection. Acta Neurochir(Wien). 2006;138(7):829–34.
Peacock WJ, Currer TH. Hydrocephalus in childhood. A study of 440 cases. S Afr Med J. 1984;66(9):323–4.
Mwang’ombe NJ, Omulo T. Ventriculoperitoneal shunt surgery and shunt infections in children with non tumour hydrocephalus at the Kenyatta National Hospital. Nairobi East Afr Med J. 2000;77(7):386–90.
Hamada SM, Ahmed H. Paediatric ventriculoperitoneal shunt—is free hand placement of ventricular catheter still acceptable? Egyptian J Neurosurg. 2015;30(3):195–8.
Aldrich EF, Harmann P. Disconnection as a cause of ventriculoperitoneal shunt malfunction in multicomponent shunt systems. Pediatr Neurosurg 1990–1991; 16(6):309–311.
Bierbrauer KS, Storrs BB, McLone DG, et al. A prospective, randomised study of shunt function and infections as a function of shunt placement. Pediatr Neurosurg. 1990–1991; 16(6):287–91.
Yung S, Chan TM. Pathophysiological changes to the peritoneal membrane during PD-related peritonitis: the role of mesothelial cells. Mediat Inflamm. 2012;2012:484167.
Kwok CK, Yue CP, Wen HL. Bilateral scrota lmigration of abdominal catheters: a rare complication of ventriculoperitoneal shunt. Surg Neurol. 1988;31:330–1.
Sathyanarayana S, Wylen EL, Baskaya MK, et al. Spontaneous bowel perforation after ventriculoperitoneal shunt surgery: case report and a review of 45 cases. Surg Neurol. 2000;54:388e96.
Berry JG, Hall MA, Sharma V, et al. Amulti-institutional, 5-year analysis of initial and multiple ventricular shunt revisions in children. Neurosurgery. 2008;62:445–53. discussion453-444
Patwardhan RV, Nanda A. Implanted ventricular shunts in the United States: the billion-dollar-a year cost of hydrocephalus treatment. Neurosurgery. 2005;56:139–44. discussion144-135
Tuli S, Drake J, Lawless J, et al. Risk factors for repeated cerebrospinal shunt failures in pediatric patients with hydrocephalus. J Neurosurg. 2000;92:31–8.
I would like to thank Dr. Mohammed A. Abdelsamea, an assistant professor at Qena Faculty of Education, South Valley University, Egypt, and a postdoctoral research associate at College of Education and Human Development, University of Minnesota, USA, for his help in editing and proofreading the English language of the final version of the manuscript.
Availability of data and materials
The data and materials of this manuscript are available for sharing.
The data collection and the scientific writing were done by the corresponding author himself. The author read and approved the final manuscript.
Ethics approval and consent to participate
An informed signed consent was taken from all the patients and guardians of patients before enrolling them into the study after approval of the ethical committee of the Faculty of Medicine, South Valley University.
The present study has been performed in accordance with the Declaration of Helsinki and after approval of Ethical Committee, Qena Faculty of Medicine, South Valley University.
Consent for publication
A written consent has been obtained from every included patients and guardians of the patients regarding publishing their details and images.
The author declares that he has no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Hamdan, A.R. Ventriculoperitoneal shunt complications: a local study at Qena University Hospital: a retrospective study. Egypt J Neurosurg 33, 8 (2018). https://doi.org/10.1186/s41984-018-0008-5
- Cerebrospinal fluid shunts
- Ventriculoperitoneal shunt
- Ventriculoperitoneal shunt complications